Application area: Normal care
Product group: Wall supply units (classic)
The all-rounder with huge potential
IV 1054 convinces customers through its almost unlimited equipment and expansion options.
Low and high voltage current, data and communications connections and medical gases are easily accessible at all times. In addition, aluminium or stainless steel support rails can accommodate all medical apparatus.
The system is particularly flexible when it comes to the media feeds, which can be routed from the back, on both sides or from the top. Depending on your requirements, IV 1054 can also be designed as a single, double or triple-channel system.
The screw-free front profiles of IV 1054 make cleaning easy and hygienic and these can also be obtained electrostatically powder-coated in an anodised design or in any RAL/NCS colour.
We did not install an integrated lighting solution for the IV 1054 in order to achieve maximum flexibility. modulux pure can be used as a matching and cost-effective enhancement as a pure lighting and/or additional supply unit.
Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation
Protection class: I
Protection type: IP 20
medical gas technology
Operating pressure of medical gas technology
Oxygen: 5 bar
Compressed air: 5 bar
Vacuum: - 0,8 bar
Media current feed: Rear, from top or side
Optional equipment rail: Above and below
Material of optional equipment rail: Stainless steel or aluminium
Number of media channels: One to three
Additional load: max. 50 kg/m
Assembly, cleaning, maintenance and repair
The medical supply units and single-bed lighting fixtures can be installed and mounted on solid or plasterboard construction walls. Customer provided substructure is required for plasterboard walls. Project-specific substructures are used for ceiling-mounted supply units.
Our products can be cleaned with conventional cleaning agents and disinfected with alcohol-free disinfectants.
Surfaces are protected by high-quality electrostatic powder coating. Non-coated parts are made of anodized aluminium or chrome-nickel steel. All plastic parts are resistant to cleaning agents and disinfectants.
The system must be maintained for the first time after 5 years and then after every 2 years. A contractual service agreement can be concluded for the maintenance work if required. More information can be found in the respective operating instructions.