Application area: Operating rooms
Product group: Bridge systems
The bridge to the OR of tomorrow
With OP 3800 modul technik provides you with a fully-integrated OR medical care system, since as well as the integrated connections for high voltage and low voltage current, medical gases and data and communications technologies, with the circulating air flow between the media bridge and ceiling, a disruption and turbulence-free laminar air supply is integrated into the system.
The media bridge is equipped according to your individual specifications and requirements. We deliver the OP 3800 connection-ready and assembled. Maintenance costs are also minimised through the rigid tubing up to the tapping points for medical gases.
You can even choose the shape of the OP 3800. In addition to the standard rectangular design, it can also be supplied as a U–shaped or L-shaped design.
Your additional advantage is that with the OP 3800 you retain your flexibility regarding any later expansion to the Hybrid-OR. Retrofitting of cables and connections for imaging procedures does not pose any problem at all.
The GW 2500 apparatus trolley is used as a carrier system for moduflex 2500 and the OP 3800 media bridge. Medical apparatus can be freely positioned along the height-adjustable consoles. The trolley is continuously braked by a manual friction brake.
In addition to GW 2500, the inner running GW 3600 can be used on the OR 3800 in order to incorporate more material into the work place.
Assembly, cleaning, maintenance and repair
The medical supply units and single-bed lighting fixtures can be installed and mounted on solid or plasterboard construction walls. Customer provided substructure is required for plasterboard walls. Project-specific substructures are used for ceiling-mounted supply units.
Our products can be cleaned with conventional cleaning agents and disinfected with alcohol-free disinfectants.
Surfaces are protected by high-quality electrostatic powder coating. Non-coated parts are made of anodized aluminium or chrome-nickel steel. All plastic parts are resistant to cleaning agents and disinfectants.
The system must be maintained for the first time after 5 years and then after every 2 years. A contractual service agreement can be concluded for the maintenance work if required. More information can be found in the respective operating instructions.